|Sensor Tip Diameter||.013''|
|Accuracy||±0.1°C in the physiological range|
|Time Constant:||0.025 seconds|
|Lead Length||5 feet|
|Sterilization Protocol||Gas ETO, Gluteraldehyde, Isoprophyl Alcohol|
ICT-4 Micro-Surface Microprobe - For use on integrated circuits and other micro specimens. 0.013'' diameter .5'' straight stainless steel shaft; slightly protruding sensor bead to facilitate temperature probing. 5'' tubular handle fits micro-manipulators. Maximum intermittent temperature 200°C, continuouse use 100°C, Time constant 0.025 Seconds. 5 foot lead. Not isolated.
Time Constant is defined as the time required to reach 63% of final temperature in liquid. An accurate reading is obtained when the reading stops changing, typically 5-10 time constants.
Maximum Temperature refers to intermittent use of probe. When probe is used for continuous measurements, maximum temperature should be reduced by 25% unless specified.